The «Where?», the «How?» and the «What For?»
The «Where?», the «How?» and the «What For?»
The goal of testing is to find the number of asbestos fibers in your sample. Our Asbestos Sampling Guide sets out the procedures for taking and preparing samples from a variety of locales.
Asbestos testing usually falls into two categories: bulk sample and air sample testing.
As testing methods, we typically use Polarized Light Microscopy (PLM) for bulk samples and Phase Contrast Microscopy (PCM) for air samples.
|Bulk Sampling||Air Sampling|
|Testing solid chunks of building material
We analyze bulk samples using Polarized Light Microscopy (PLM). Polarized light is a contrast-enhancing technique that improves the quality of the image obtained when examining minerals. We exploit the optical properties specific to asbestos and reveal detailed information concerning the structure and composition of this hazardous mineral.(1)
Please note: If polarized light microscopy proves inconclusive, we have the option of using Transmission Electron Microscopy (TEM), which requires a minimun turnaround time of 24 hours. If we determine that your sample requires TEM, we’ll call you immediately to let you know.
|Testing indoor air
Air sampling is used to find asbestos that might be floating in the air.
The majority of air tests are for:
We analyze air samples using Phase Contrast Microscopy (PCM). Similar to PLM, PCM uses the contrasting and light-phase properties of the substance being tested. Phase contrast microscopy takes advantage of minute refractive index differences within cellular components and between unstained cells and their surrounding aqueous medium to produce contrast in these and similar transparent specimens.(2)
Air samples are collected by running air through a filter contained within a cassette.
Please note: A major challenge with air sampling is receiving a cassette overloaded with dirt and contaminants. The air test has to be run in a clean, dry area. The pump opening should be facing down, 5-6 feet (1.5 – 2m) above the floor surface.
How To Collect A Bulk Sample – The Right Way
The basic steps to collecting a bulk sample are:
- Use personal protective equipment (including dust mask, safety glasses, and gloves).
- Collect the material, put it in a Ziploc bag and seal.
- Label the Ziploc bag. (ex: “B1 = living room ceiling – popcorn ceiling”).
- Place all Ziploc bags in one larger Ziploc bag and label it with the address where the sample was taken.
Examples of Bulk Samples
A small piece of solid building material is all that’s required for bulk sampling, typically the size of a dime or the tip of your pinky finger.
Below is a sampling guide to the most common building materials where asbestos is found:
|Type||How to collect it|
|Drywall Compound||Take a sample from a wall corner where drywall mud is most concentrated.|
|Vinyl Floor Tiles (VCT Floor Tiles)||Pick a corner and use a hook-knife or small prybar to cut off a small piece.|
|Linoleum Flooring||Use a utility knife to cut off a small piece from the corner.|
|Silicone Around Windows and Sliding Doors||Use a utility knife. Cut once on the window side, and once along the trim side to get a small piece (an inch long works great).|
|Textured Popcorn Ceiling||You’ll want to avoid just scraping the ceiling because the popcorn texture will break into pieces and make a mess (remember it might also be contaminated with asbestos).
Instead, stick a 6-inch piece of duct tape to the ceiling. Using a trowel you can break off the popcorn-texture that the duct tape is covering. The texture will stick to the duct-tape, and then you can pull it off without a mess. Store your piece of tape in a Ziploc and it’s ready for delivery.
|Textured Walls||Use a knife to cut a piece off the wall. The entire wall will be textured, so you don’t have to only cut from the corner.|
|Plaster||Use a knife to cut a piece off the wall, or you can use a hammer on the outside corner to break off a piece.|
|Thermal Insulation Around Boiler||Use a utility knife to cut off a 2×2 square (inches), and put some tape on the hole you just made to protect the integrity of your insulation. The size of the sample should be around 2×2 inches.|
|Attic and Wall Insulation||Use a utility knife to cut off a 2×2 inch square.|
|Roofing Materials||On a flat roof, there may be asbestos in the tar, or in the paper underneath.
On a cedar-shakes shingle roof, most of them have asbestos in the paper barrier underneath the shingles.
On asphalt shingles, you have to test the paper underneath the shingle for asbestos. Usually there isn’t asbestos in the shingle itself.
If you or someone else is living in a house where samples need collecting, look for a discrete area to retrieve the sample, such as:
- Behind the stove
- Behind the fridge
- Behind the toilet
- In a closet
Anywhere that won’t be seen is the best place for a sample when someone is still living in the building.
Use safety gear when collecting the sample and place it in a plastic Ziploc bag, labelling each bag as you go.
Here’s an example of how you might label your bulk samples:
- B1 = Living room ceiling – popcorn ceiling
- B2 = Dining room corner – drywall compound
- B3 = Bathroom floor tiles – linoleum
How Many Samples Do I Need?
|Square Footage Of Surface||Samples Required|
|Less than 968 sq. ft.||minimum 3 samples|
|968-4,843 sq. ft.||minimum 5 samples|
|4,843+ sq. ft.||minimum 7 samples|
Once the samples are labelled and in Ziploc bags, put all your samples into one larger bag and label it with the address. Download the Chain of Custody form, fill in the information on location, sample type and preferred results delivery method and contact our office at 416.688.2757 for sample pick-up.
Save Hours Delivering Asbestos Samples Using Our Automatic Asbestos Sampling Service
How To Properly
Collect An Air Sample
Collect An Air Sample
The simple steps to collecting an air sample are:
- Place your air pump with the cassette inside each of the rooms you are testing.
- Run your air pump with the cassette inside. You’ll need 2,400 litres for a clearance test, and 480 litres for personal air monitoring.
- Remove the cassette from the pump & label it (ex: “B1 = main floor living room, B2 = 2nd floor, washroom on north side.”)
- Place all cassettes in a Ziploc bag and write the address location on it.
- Fill out our Chain of Custody form and call us at 416.688.2757 for pick-up.
Running air at the right speed is vital. While you can get a sample quicker by turning up the pump, you can actually collapse the filter in your cassette if it’s too fast. NIOSH 7400 requires 2.5-16L/minute as the proper airflow.
You wouldn’t pump only 2.5L/minute for most air samples (it would take 16 hours to get 2,400 litres of air!). For that type of volume, you’d want to run the pump at 16L/minute (2.5 hours to finish).
Many regulators can go up to 20 L/min, but we’ve found it to be just too much air for the filters, and sometimes filters will collapse (setting you back to restart the test entirely). So try not to run the pump any higher than 16L/ min.
You need to track how long the pump was running, and what speed – down to the minute! You’ll need to write this data down on the Chain of Custody form when delivering the samples.
Once the pump has run enough air through the filter, put the cassette in a plastic Ziploc bag and label it accordingly. Here’s an example of how you might label your air samples:
- B1 = Living room
- B2 = Dining room
- B3 = Upstairs bathroom
Put all your samples together in a single larger Ziploc, label the bag with the building address, and call us for sample pick up. Make sure the Chain of Custody form is in the larger bag so we know what each sample bag contains.
Air sampling requires a special air pump. You need to run a certain amount of air through the cassette to get a proper reading. Some tests require 2,400 litres of air, while others only require 400.
Air samples are collected in a specialized cassette, which is a plastic tube with a filter in the bottom.
When air runs through the cassette, particles get stuck in the filter. When the filter comes to us, we analyze the particulate matter trapped in that filter and deliver you the analysis results of that sample.
Not An Expert In Sampling?
There are a lot of variables to consider to get a sample taken correctly. That’s why all air tests should be performed by a trained PCM technician (Phase Contrast Microscopy, NIOSH 582 equivalent certification).
If you need samples taken, but don’t have the expertise, take advantage of our automatic sampling service. A trained technician will come to your site, take the sample(s) and return to the lab, and deliver your test results by phone, FAX or email as requested. Automatic sampling is also available for bulk samples.